Available exclusively worldwide only on the OpenBCI® store.
About Conscious Labs
Conscious Labs has been working since 2014 on taking neuroscience out of the lab and into the wild. They bring the technology, the team and the expertise to integrate EEG sensors into head wearables and hearables.
We are very excited to partner with Conscious Labs to offer this affordable, high-quality biosensing kit to our community. The ThinkPulse™ kit offers research-grade dry, active electrode EEG sensors at an unprecedented price point and comfort level.
ThinkPulse™ is a product of Conscious Labs (France). Patents issued and pending. Made in France.
The ThinkPulse™ system is an add-on to the Ultracortex EEG Headset plus Cyton Board This kit has been designed and its compatibility tested for the 8 and 16 channel Ultracortex with OpenBCI Cyton/CytonDaisy boards. For more compatibility and customization options as well as general product questions, please contact us firstname.lastname@example.org..
ThinkPulse Starter Kit $349.99 Contents
Note, the replacement set $99.99 is just for those who want backup electrode tips. The Starter Kit $349.99 includes the necessary electrodes and electrode tips.
How to Use the ThinkPulse Starter Kit with the Ultracortex Headset
Below is the step-by-step tutorial by Conscious Labs on how to install the ThinkPulse Starter Kit on the Ultracortex. If you are doing this for the first time, allow yourself 20 minutes to put it together. You may briefly need a soldering iron and soldering wire at Step 3. If you do not have a soldering iron, you can still assemble the ThinkPulse! Keep reading for the solder-free option.
Step 1: Prep Your Ultracortex Mark IV EEG Headset
Step 2: Install the ThinkPulse™ sensors on the Ultracortex
Step 3: Assemble the ThinkPulse™ Power Supply Board
The assembly video shows how to install the power supply board to a post-2018 Cyton and a pre-2018 Cyton.
Some early ThinkPulse units have a hardware bug. The fix is a straightforward, quick solder job. The first minute of the video explains how to do it.
ThinkPulse kits that shipped post-January 2021 are fixed and do not require soldering. If you need help or have any questions, email email@example.com
Step 4: Connect and organize the cables
Step 5 (For 16-channel kit) Connect and organize the cables
Using the ThinkPulse™ sensors with OpenBCI Cyton Board
The following information applies specifically to Cyton and Cyton+Daisy Boards. If you are seeing
RAILED in the GUI, try the following tips:
- Disconnect the BIAS earclip from the Cyton board.
- Go to the Hardware Settings (button is found above Time Series widget) and turn BIAS and SRB1 to
- Keep SRB2
- For every channel, set the GAIN to '8' or less.
- Below, you will find a screenshot of what this should look like in the GUI Hardware Settings. If you are communicating directly with the Cyton, the command correlating to this picture would be
- Pressing the
Savebutton at the bottom of the Hardware Settings will allow you to store this configuration as a file.
- Next time you want to use the Cyton with ThinkPulse Electrodes, open Hardware Settings to
Sendchannel settings. If you are using BrainFlow outside of the GUI, simply send these commands after instantiating the Board.
Technical Specs and Data Sheet
|Electrode Height||7 mm|
|External Diameter||19 mm|
|Bio-contact Area Diameter||15 mm|
|Polymer Dielectric Resistivity||3.6 Ohm.cm|
|Through Polymer Resistance||<70 Ohm|
|Typical Consumption||5 mW|
|Active amplification Gain||0 dB|
When using the ThinkPulse Active Electrodes with Cyton, make sure to set the correct channel settings. This should resolve most issues. If you still have issues, you can reach out on the OpenBCI Forum or send us an email at firstname.lastname@example.org.
You can also refer to this resolved troubleshooting thread in the OpenBCI Technical Forum.
Non-medical Use Disclaimer
The ThinkPulse™ Starter Kit and the ThinkPulse™ Electrode Set are not medical devices and are not intended for medical diagnosis. These products are provided to you "as is." OpenBCI, Inc. provides the above product under the following conditions: This evaluation kit is intended for use for ENGINEERING DEVELOPMENT, DEMONSTRATION, OR EVALUATION PURPOSES ONLY.